<!DOCTYPE html>
<html>

<head>
    <meta charset="utf-8">
    <title>Certificate of Instruments TMS50i</title>

    <style>
        @import url('https://fonts.googleapis.com/css2?family=Playfair+Display:ital,wght@0,400..900;1,400..900&display=swap');

        @page {
            margin: 15mm;
        }

        body {
            font-style: italic;
            margin: 0;
            padding: 0;
            color: #000;
            font-size: 13px;
        }

        .container {
            width: 100%;
        }

        .doc-number {
            font-style: normal;
            padding: 0;
            margin: 0;
            text-align: right;
            font-size: 12px;
        }


        .title {
            margin: 30px 0px;
            text-align: center;
            font-style: italic;
            text-decoration: underline;
            font-size: 20px;
        }

        h4 {
            margin: 0 0;
            padding: 0;
            font-size: 14px;
        }

        .customer-table {
            width: 100%;
            border-collapse: collapse;
        }

        .customer-table td {
            padding: 3px 0;
            vertical-align: top;
        }

        .customer-table td.label {
            width: 180px;
        }

        .box {
            border: 1px solid #000;
            padding: 12px;
            margin-top: 20px;
        }

        /* .box h4 {
            margin: 0;
            margin-top
        } */

        .result-table {
            width: 100%;
            border-collapse: collapse;
            margin-top: 10px;
            font-size: 13px;
        }

        .result-table th,
        .result-table td {
            border: 1px solid #000000;
            padding: 5px 6px;
            text-align: center;
        }

        .result-table td.param {
            text-align: left;
        }

        .conclusion {
            margin-top: 15px;
        }

        .footer {
            margin-top: 20px;
            width: 250px;
        }

        .date {
            margin-bottom: 70px;
            font-style: italic;
        }

        .name {
            /* font-weight: bold; */
            font-style: italic;
            border-bottom: 1px solid #000;
            display: inline-block;
            width: 63%;
            padding-bottom: 2px;
        }

        .position {
            margin-top: 5px;
            font-style: italic;
        }

        /* Bersihkan float agar elemen setelahnya tidak berantakan */
        .clearfix::after {
            content: "";
            clear: both;
            display: table;
        }
    </style>
</head>

<body>
    <div class="container">

        <div class="doc-number">
            Nomor: FO.III.12/04.00/2019
        </div>

        <div class="title">
            <h3>Certificate of Instrument Calibration</h3>
        </div>

        <div class="section">
            <table class="customer-table">
                <tr>
                    <td class="label">
                        <h4>Customer Data</h4>
                    </td>
                    <td></td>
                </tr>
                <tr>
                    <td class="label">
                        Laboratory / Hospital
                    </td>
                    <td>
                        : Trisensa Diagnostic Center
                    </td>
                </tr>
                <tr>
                    <td class="label">
                        Address
                    </td>
                    <td>
                        : Jl. Taman AIS Yasin No.5, Embong Kaliasin, Kec. Genteng, Surabaya 60271
                    </td>
                </tr>
                <tr>
                    <td class="label">
                        Instrument
                    </td>
                    <td>
                        : Clinical Chemistry Analyzer TMS 50i Superior
                    </td>
                </tr>
                <tr>
                    <td class="label">
                        Serial Number
                    </td>
                    <td>
                        : 5303150816
                    </td>
                </tr>
                <tr>
                    <td class="label">
                        Date of Calibration
                    </td>
                    <td>
                        : January 09, 2026
                    </td>
                </tr>
                <tr>
                    <td class="label">
                        Valid Until
                    </td>
                    <td>
                        : January 09, 2027
                    </td>
                </tr>
            </table>

        </div>

        <div class="box">
            <h4>Required Material</h4>
            Aquadibidestilata

            <h4>Procedure</h4>
            Running 60 samples (Aquadibidestilata) for 12 Wavelength
            <br>
            (340, 380, 405, 450, 505, 546, 570, 600, 660, 700, 750 and 800 nm)

            <h4>Requirement</h4>
            Expected Result : (-20) - (+20)

            <h4>Result</h4>

            <table class="result-table">
                <tr>
                    <th>Parameters</th>
                    <th>Value</th>
                    <th>Range</th>
                    <th>Unit</th>
                    <th>Result</th>
                </tr>
                <tr>
                    <td class='param'>Incubation Module</td>
                    <td>37.0</td>
                    <td>36.7 - 37.3</td>
                    <td>°C</td>
                    <td>Pass</td>
                </tr>
                <tr>
                    <td class='param'>Reagent Compartment</td>
                    <td>5.0</td>
                    <td>2.0 - 8.0</td>
                    <td>°C</td>
                    <td>Pass</td>
                </tr>
                <tr>
                    <td class='param'>Power 24V Voltage</td>
                    <td>23.94</td>
                    <td>20 - 30</td>
                    <td>Volt</td>
                    <td>Pass</td>
                </tr>
                <tr>
                    <td class='param'>Power 12 V Voltage</td>
                    <td>11.87</td>
                    <td>11.0 - 14.0</td>
                    <td>Volt</td>
                    <td>Pass</td>
                </tr>
                <tr>
                    <td class='param'>Vacuum Pressure Check</td>
                    <td>-23.00</td>
                    <td>-24.00 - -34.00</td>
                    <td>KPa</td>
                    <td>Pass</td>
                </tr>
                <tr>
                    <td class='param'>DCF Diagnosis</td>
                    <td>0.96</td>
                    <td>
                        < 1.5%</td> <td>%
                    </td>
                    <td>Pass</td>
                </tr>
                <tr>
                    <td class='param'>Substrate Background</td>
                    <td>48</td>
                    <td>SD < 200</td> <td>N/A</td>
                    <td>Pass</td>
                </tr>
                <tr>
                    <td class='param'>System Repeatability</td>
                    <td>0.12%</td>
                    <td>CV ≤ 1.5%</td>
                    <td>%</td>
                    <td>Pass</td>
                </tr>
                <tr>
                    <td class='param'>Dark Current Count</td>
                    <td>80.2</td>
                    <td>0 - 350</td>
                    <td>N/A</td>
                    <td>Pass</td>
                </tr>
            </table>

            <div class="conclusion">
                <h4>Conclusion</h4>
                Instrument Chemiluminescence Immunoassay System Mindray CL-900i have been already calibrated.
                    The result of calibration is good.
            </div>
        </div>

        <div class="clearfix">
            <div class="footer">
                <div class="date">Jakarta, January 23, 2026</div>
                <div class="name">Adhitya Pranata Putra</div>
                <div class="position">Technical Support Manager</div>
            </div>
        </div>

    </div>
</body>

</html>
